1、The machine positioning accuracy, reliable performance, can be adapted to the general radiography. The use of bridge rectifier combined x-ray generator, which can improve the quality of radiography.
2、Adopt grid, the image is more clear, especially for the post-animal shooting.
3、Configuration filament regulator circuit, the parameters are more stable.
4、Designed for animals radiography, the operation is more convenient and simple.
二、The main technical parameters
1、Power supply conditions and operating mode
Power supply voltage: AC 220V ± 22V;
Power frequency: 50Hz ± 0.5Hz;
Power capacity: ≥ 8KVA;
Power supply allows maximum internal resistance: 1Ω Operation mode: intermittent loading continuous operation
2、The highest rated capacity is shown in Table 1(Table 1. Maximum rated capacity
Tube current（mA）Tube voltage（kV）Time（s）
15 90 6.3
30 90 6.3
60 90 4.0
100 80 3.2
Tube voltage: 50-90kV
Tube current: 15,30,60,100 mA total of 4 shifts;
Time: 0.08s-6.3 s, a total of 19 shifts, according to R10 'coefficient selected.
4、Maximum output power:
( 80kV 100mA 0.1S ) 5.92KVA.
5、Nominal electric power
90kV 60mA 0.1S 4.00KVA.
6、Input power: 5.92KVA.
When the x-ray generator window is down, the distance between the focus and the film is 1000mm;
The x-ray generator has a rotation angle of ± 90o around the vertical axis;